View and Download ABB Advance Optima Uras 14 service manual online. Advance Optima Uras 14 Measuring Instruments pdf manual download. Manuals and User Guides for ABB Advance Optima Uras We have 1 ABB Advance Optima Uras 14 manual available for free PDF download: Service Manual. Spare Parts for ABB & Siemens Gas Analyzer & Sample Handling Systems is available. Home» Products» ABB Gas Analyzer Spare Parts» Module Uras
|Published (Last):||1 June 2005|
|PDF File Size:||11.70 Mb|
|ePub File Size:||2.64 Mb|
|Price:||Free* [*Free Regsitration Required]|
Module components Chapter 4: No measurement signal Indication No measurement signal when sample gas concentration changes. In that case turn of the compensation temperature detector. Thermal link Accessories Multimeter, screwdriver Resistance Electrical value: The main frame always has xbb apertures installed and these are moved back and Structure forth horizontally by a screw-type drive mechanism.
Module components Overview This chapter describes the individual assemblies and components. Sample cell Location in module The sample cell is installed between the main frame with modulator and the infrared detector or calibration unit. Calibration Overview Introduction This chapter describes how to prepare for initial calibration of the analyzer module and describes how to perform the initial calibration, postlinearization and cross- sensitivity corrections.
Structure The board is connected to the IR module circuit board. Update mode Cyclical or query urad raw jras.
URAS 14 Gas Analyzer Ao ABB Ao ** | eBay
Physical measurement value Explanation: Change modulator Note The modulator can be changed without removing the analyzer module. Change Emitter Insert Change emitter insert Note The emitter inserts can be changed without removing the analyzer module.
The circuit board contains two heat resistors. Unstable Measurement Value Reading Unstable measurement value reading Indication Measurement signal indicating non-reproducible shifts of varying amplitudes. Special accessories are described where necessary. Chapter contents In this chapter you will find the following information: Urqs infrared detector consists of the following components: Step Action Turn off the analyzer power supply.
Pressure Detector Pressure detector General The pressure detector pressure 1 is always installed in the analyzer module and thus is always configured. After calibration messages 1, 2, 4, 8 or 16 may appear. Physical measurement value; low-pass Explanation: Remove three screws 1. Page 27 Module versions plastic tubing connectionscontinued Figure Analysatormodul Uras 14 Pneumatikmodul Module versions: The modulator consists of: Hardware configuration, continued Detector Make sure the proper closing disk is installed.
Module Uras 14
Sample cell, continued Figure Sample cells NL 0. Add to Watch list. This item may or may not be in its original packaging. Disconnect the gas lines from the analyzer module ports. Calibration cell Location in module The calibration cell is a option to be mounted in the calibration unit. Electrical connections What electrical The following connections are always present: Page 53 IR module circuit board, continued Figure No status messages Status messages Details Possible cause Measures Improper sample gas preparation – Check the cooler – Check dew points of sample gas and, if yras, zero gas Sample components in zero gas Replace zero gas Inadequate purge time prior to zero Page Hardware configuration, continued Detector Make sure the proper closing disk is installed.
Page Urae sample cell, Continued Figure Removing sample cell with calibration unit Chapter 6: IR detectors The example here is for the measurement of CO, NO, SO and O Uras detector configuration Note that while the oxygen sensor is not part of the Uras module it must always be configured together with the analyzer module. Configuration Phase Alignment, continued Execution After the phase alignment function is activated the process is menu-driven.
ABB Advance Optima Uras 14 Manuals
For this reason, variable measurement ranges are not possible. Module components Emitter Modulator Ask a question about this product Call Page Subject to technical changes Printed in the Udas.
Temperature control board terminal strip X2 Remove the thermal link from the terminal strip and urxs the thermal link continuity. Structure A small drive motor with a drive pin moves the slide on which the calibration cell is mounted. Emitter Emitter Location in Module The emitter inserts are attached to the modulator receiver disk aluminum block. Between yellow and green approx.
Sample Cell Sample cell Location in module The sample cell is installed between the main frame with modulator and the infrared detector or calibration unit. Calibration unit Accessories Multimeter, PC with test- and calibration software The upper, lower and lateral travel paths of the calibration cell slide must be free of Visual check interference by cables and tubes.
Figure Ex module connections Gas connectors Electr. Page Temperature detectors, continued Component configuration Component Component name e. There are two side-mounted terminal strips for two ribbon cable connections to the optical components. Optical Filter Optical filter Location in module Depending on the module layout and measurement task, the optical filters are located at the following points: